CloverSnare 4-Loop Recall Issued Over Pulmonary Embolism Risk
Following at least four reports of injuries, federal regulators are warning that a device used to retrieve items from the cardiovascular system may lead to the very injuries it was designed to prevent.
The FDA gave a Cook CloverSnare 4-Loop Vascular Retrieval Snare recall its most serious designation this week, indicating that continued use of the device may pose a risk of serious injury or death, as the tip of the snare may be prone to break off during use.
The action was given a Class I recall designation shortly after the company announced the recall in a press release on August 26.
Did You Know?
Millions of Philips CPAP Machines Recalled
Philips DreamStation, CPAP and BiPAP machines sold in recent years may pose a risk of cancer, lung damage and other injuries.Learn More
The Cook CloverSnare 4-Loop Vascular Retrieval Snare is used to manipulate and retrieve items in the cardiovascular system, usually blood vessels. It is often used to maneuver temporary implanted devices, such as vena cava filters, broken devices, guide wires, coils, balloons and catheters.
The recall warns users that the snare loop may separate from the shaft, possibly causing embolization, or blocking of the blood vessels, with the device fragments. If this happens, the snare loop may travel through the vascular system and become stuck in other organs, like the heart and the lungs.
The FDA warns additional interventions may be necessary to retrieve the snare loop if separated from the patient. This will also cause the device to stop working.
The problems caused by the recalled device may cause serious side effects, health concerns and may result in death.
The recall affects Cook CloverSnare 4-Loop Vascular Retrieval Snarea with model number VRS-6.0-9.0 and manufacturing dates of August 2012 through August 2013. The devices were distributed from March 8, 2013, through July 1, 2014. Nearly 700 were distributed nationwide.
The snares are manufactured by Cook Inc. In Bloomington, Indiana.
Cook Medical distributed an Urgent Medical Device Recall letter to customers on July 17 informing them of the problem and actions which need to be taken. Cook also instructed customers to separate the affected unused products from inventory, collect the defective used devices and return all unused products to the company.
"*" indicates required fields
More Top Stories
The U.S. JPML has consolidated all Tepezza lawsuits over hearing loss before one Northern Illinois federal judge for coordinated pretrial proceedings.
Bard claims two cases selected for the third and fourth bellwether trials are no longer representative of the litigation due to the plaintiffs' worsening injuries and need for additional surgeries due to their failed hernia mesh products.
More than 775 Exactech lawsuits have been filed in federal and state courts as parties work toward a plan for bellwether early test trials.