Penumbra Coil Recall: Aneurysm Device May Migrate in Brain
The Penumbra Coil recall was announced by the FDA earlier this week. Penumbra Inc. notified its customers in March and the FDA has classified it as a Class 1 recall, meaning that the defective medical device has a reasonable probability of causing severe injury or death.
The Penumbra Coil 400 is a small platinum coil that is surgically implanted into a brain aneurysm via blood vessels leading to the brain. If implanted correctly, a blood clot forms around the coil, occluding the aneurysm and keeping the blood vessel from bursting.
Did You Know?
Millions of Philips CPAP Machines Recalled
Philips DreamStation, CPAP and BiPAP machines sold in recent years may pose a risk of cancer, lung damage and other injuries.Learn More
However, the FDA warns that the Penumbra Coil system includes a tool for implanting the coil that is defective. A pull wire on the tool can slip out of place, resulting in the coil detaching from the tool prematurely. This allows the Penumbra Coil to migrate in the brain.
The recall affects 267 lots of the Penumbra Coil 400; from lot number F17211 to F18553. Penumbra Inc. has stated that it sent a letter to customers and distributors to return the products by mail and all of them have been accounted for and returned.
The Penumbra Coil was sold both inside and outside of the United States.
Any customers with questions can call Penumbra, Inc. at 510-748-3224. Any health care professionals or consumers who have experienced an adverse event due to Penumbra Coil side effects should report it through the FDA MedWatch Safety Information and Adverse Event Reporting Program.
LannyAugust 23, 2022 at 11:12 am
I've had the implant and has suffered from this procedure
"*" indicates required fields
More Top Stories
A Wegovy gastroparesis lawsuit blames the weight loss drug for a stomach paralysis problems which left a woman with permanent injuries.
Uber faces a lawsuit from four passengers who say they were sexually assaulted by drivers, due to the company's lack of security measures and focus on passenger safety.
A Bard PowerPort lawsuit claims the defective design of the port catheter led to a woman developing a severe infection and needing to have the implant surgically removed.