FDA Warning Letter Sent to Fresenius Over Dialysis Product Design
Federal regulators have issued a warning letter to Fresenius Medical Care, the nation’s largest provider of dialysis services and products, indicating that it failed to adequately study artificial kidneys used in its dialysis machines.
The company announced the FDA warning letter in a press release to investors on March 14, indicating that the FDA criticized the company for not making sure that the dialyzers that were manufactured adequately matched the actual design specifications.
Specifically, Fresenius “did not conduct adequate design verification studies of its electron beam sterilized polysulfone dialyzers,” which were manufactured at its Ogden, Utah, facility.
Did You Know?
Millions of Philips CPAP Machines Recalled
Philips DreamStation, CPAP and BiPAP machines sold in recent years may pose a risk of cancer, lung damage and other injuries.Learn More
Although the company received clearance for the dialyzers in December 2000, the company indicates that the design validation for the artificial kidneys is incomplete.
Fresenius has indicated that it will work with the FDA to address the agency’s concerns as soon as possible. The letter did not call for a Fresenius artificial kidney recall.
Fresenius Problems with Dialysis Products
The warning letter comes as Fresenius is under close scrutiny following a NaturaLtye and GranuFlo recall, which the FDA forced the company to issue after the company warned about an increased risk of cardiac arrest and sudden death during dialysis treatments.
Questions have been raised about whether Fresenius withheld information about the risk of problems during dialysis treatment with Granuflo and NaturaLyte. Although the company sent an internal memo to it’s own clinics in late 2011, they failed to inform medical providers at competing clinics that used their dialysis products about the importance of monitoring patients until March 2012. The FDA classified this warning as a recall.
An increasing number of Fresenius dialysis lawsuits have been filed nationwide on behalf of individuals who suffered heart attacks or lost loved ones to heart problems after dialysis treatment with NaturaLyte and GranuFlo.
All of the lawsuits involve similar allegation, claiming that the company failed to provide adequate warnings about the risk of complications during Granuflu or Naturalyte dialysis treatment, including the importance of monitoring patients’ bicarbonate levels to avoid a risk of sudden cardiac arrest or death.
"*" indicates required fields
More Top Stories
A new report highlights how many women and families feel left out of Camp Lejeune settlement negotiations after suffering repeated miscarriages they say were caused by miscarriages on the North Carolina military base.
A Bard Infuse-A-Port lawsuit claims a piece of a failed port catheter broke off, causing a woman to suffer a pulmonary embolism which has resulted in fragments of the device remaining in her heart.
A Wegovy gastroparesis lawsuit blames the weight loss drug for a stomach paralysis problems which left a woman with permanent injuries.