GE Anesthesia System Problems May Result in Overdelivery: Warning
GE Healthcare has announced a voluntary field correction action for two anesthesia delivery systems, due to potential software problems that may cause the machines to over deliver medication, posing serious and potentially life-threatening health risk for patients.
An Avance, Aisys and Avance CS2 anesthesia delivery systems field action was announced by the FDA on October 15, after the manufacturer recognized the software programs may allow over-delivery of medication when using the Pressure Control Ventilation-Volume Guarantee Mode (PCV-VG). To date no injuries or incidents have been reported.
The systems are used to deliver anesthesia to patients and are controlled by a software program that allows the user to input a particular amount of fluid to be disbursed. The affected systems may fail to deliver the correct amount of anesthesia fluid if the user transitions from mechanical PCV-VG mode to manual ventilation because changing the mode will also change the set tidal volume of anesthesia to be delivered.
Did You Know?
Millions of Philips CPAP Machines Recalled
Philips DreamStation, CPAP and BiPAP machines sold in recent years may pose a risk of cancer, lung damage and other injuries.Learn More
Over delivery of anesthesia may cause moderate to severe health consequences such as nausea, vomiting, hallucinations, and under certain circumstances cause reduction in blood pressure. In more severe circumstances patients may suffer from dementia, mental impairment, and even death.
The field action consists of Avance, Aisys and Avance CS2 anesthesia delivery systems with 8.00 or 10.00 software programs manufactured by GE Healthcare. The systems were manufactured between February 2011 and July 2013 where they were distributed for sale to various healthcare providers.
GE announced that users should not adjust the ventilator tidal volume while in the manual ventilation mode to avoid over delivery of anesthesia medication.
GE Healthcare initiated its voluntary field action by sending an Urgent Medical Device Correction letter alerting the customers of the concerns and instructions on how to avoid the issue. The letter also contains instructions on how to contact GE Healthcare to correct the systems for free. Customers with affected devices should contact GE Healthcare Customer Service center at 1-800-345-2700 or visit them online at gehealthcare.com. Customers may also contact the FDA at 1-800-FDA-1088 or 1-888-INFO-FDA for further questions.
Get more articles like this sent directly to your inbox.
"*" indicates required fields
JaniceDecember 15, 2020 at 12:20 pm
Hi this machine was used on me i had 2 blood clots to my lungs who can I contact for help
More Top Stories
A federal magistrate judge has ordered Covidien to turn over a large collection of hernia mesh complaint files to plaintiffs as the parties prepare to select potential bellwether test cases for 2025.
Four federal judges have ruled that the second group of Camp Lejeune lawsuits to be prepared for bellwether trials will involve claims of prostate cancer, kidney disease, lung cancer, liver cancer and breast cancer.
A federal judge has scheduled the initial status conference for coordinated GLP-1 RA drug stomach paralysis lawsuit pretrial proceedings for March 14.