Hamilton Ventilator Recall Issued Over Short Circuit, Capacitor Leak Risks
More than 1,000 Hamilton ventilators are being recalled, due to a risk that the devices may suddenly short circuit and fail, which could put patients at risk of severe injury and death.
The U.S. Food and Drug Administration (FDA) announced the Hamilton ventilator recall on October 18, indicating the capacitors may leak electrolyte fluid onto the device’s control board, which can result in a sudden short circuit event and cause unexpected loss of breathing assistance to patients.
Healthcare professionals use the mechanical ventilators to provide breathing support to newborns, infants, children, and adult patients while they are transported in and out of a hospital setting. While no injuries have been reported in relation to the recalled ventilators, the interrupted or complete lack of air therapy treatment may lead to serious and potentially fatal injuries.
Hamilton Ventilator Short Circuit Recall
The FDA has given the Hamilton ventilator recall a Class I designation, warning medical professionals that patients face an increased risk of sustaining severe injuries or dying if they continue using the devices.
According to the recall, leaking ventilator capacitors can short circuit the control board, which can switch the device into an “Ambient State,” causing the ventilator to suddenly stop working and fail.
In the event of a ventilator failure, patients will require immediate action from a medical professional and need an alternative device, or may require a manual resuscitation device to supply them with air.
Did You Know?
Millions of Philips CPAP Machines Recalled
Philips DreamStation, CPAP and BiPAP machines sold in recent years may pose a risk of cancer, lung damage and other injuries.Learn More
The recall impacts approximately 1,468 Hamilton ventilators distributed from July 28, 2020 until August 8, 2020 and includes the following models:
- Hamilton-C1, Portable Intensive Care Ventilator, REF: 161001
- Hamilton-T1, Intensive Care and Transport Ventilator, REF: 161006
- Hamilton-T1, Intensive Care and Transport Ventilator, REF: 161009
- Hamilton-MR1, Intensive Care Ventilator, REF: 161010
- Hamilton C1, T1, and MR1 models, Control Board Spare Part, REF: MSP161502/12
- Hamilton-T1, ESM Shielding Set Spare Part, REF: MSP161339/02
Hamilton issued an urgent medical device correction letter to healthcare professionals on July 23, requesting them to provide patients with an alternative ventilation method. The letter also provided instructions on how to exit the device’s “Ambient State,” and service the ventilator.
The ventilators can be returned to patient use after they are serviced and pass the software test requirements. For more information on the recall, consumers may contact Hamilton Medical Inc. by phone at 1-800-426-6331, or email firstname.lastname@example.org.
Consumers are urged to report any problems they experienced after using the recalled ventilators to the MedWatch Adverse Event Reporting program.
"*" indicates required fields
More Top Stories
Although Suboxone settlements have been paid to resolve antitrust violations, users who suffered damages due to tooth decay from Suboxone film must pursue individual product liability lawsuits
With thousands of Bard hernia mesh lawsuits pending in the federal court system, a fourth bellwether trial will be held in the spring, involving allegations that defects with Bard 3DMax caused painful and permanent injuries.
A Tepezza hearing loss lawsuit accuses the manufacturer of failing to warn doctors to conduct hearing tests, which could have helped a woman avoid permanent hearing damage.